Interrupt Harmful ElectroMagnetic Radiation (EMR)
Oscilloscopic Comparisons to the SRT-3 Q-Link
Q-Link’s pendant has been touted for its ability to transform harmful EMR (ElectroMagnetic Radiation) into positive energy, supposedly enabling its wearer to stay energized and focused. Q-Link is a passive device, which when viewed from the front appears to have a simple coil of copper wire wound around it center. Rhythmedics instruments are active devices emitting Fibonacci based Bio-Pulses that actively interrupt harmful EMR. NuTesla acquired a new SRT-3 Q-Link and using NuTesla’s state-of-the-art laboratory conducted a series of tests comparing the measurable and viewable affects of both the Q-Link and Rhythmedics to disrupt EMR from three common household electrical products; A) 5.8 GHz Cordless Phone, B) 12 volt Plug-in Wall Transformer used to recharge electronic devices, and C) desktop fluorescent lamp’s power supply. Three images are displayed for each electrical product tested. The first image shows the raw EMR source, the second shows the EMR with the Q-Link between the probe and source, and the third shows the EMR with the Rhythmedics Instrument between the probe and source. In every case the Q-Link shows no affect on the EMR, whereas the Rhythmedics Instrument clearly interrupts the EMR source.
1. CORDLESS PHONE EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
2. WALL TRANSFORMER EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
3. FLUORESCENT LAMP POWER SUPPLY EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
Notes: All measurements were made with a Tektronix TDS 3034 Digital Phosphor Oscilloscope. A 10 milliHenry ferrite core inductor with a 10,000 ohm termination resistor was used as the probe. Measurements were made at a distance of 1 inch from the EMR source to eliminate ambient EMR sources. Once a raw EMR signal was captured on the digital oscilloscope, the SRT-3 Q-Link was inserted between the EMR source and the probe and the second image was captured. Then the Rhythmedics Instrument was placed between the EMR source and probe and the third image was captured. All settings for the Oscilloscope are displayed on its screen.
The images above clearly show the Q-Link makes no observable changes to the EMR from common household sources. Whereas the images clearly show the Rhythmedics Bio-Pulses interrupts the continuous EMR. This is critical as studies of EMR have shown it is necessary to disrupt the continuity of the EMR stream in order to prevent that aspect of it from adversely affecting the health of humans and animals. All Rhythmedics Instruments create Bio-Pulses that effectively interrupts harmful EMR as shown above.